Thank you for the opportunity to speak today on behalf of the National Center for Health Research. I am Dr. Danielle Shapiro, I am a physician and senior fellow. Our research center scrutinizes scientific and medical data and provides objective health information to patients, providers and policy makers. Those are the perspectives I bring with me today. We do not accept funding from device companies and therefore I have no conflicts of interest.
Three to 5 % of the US population are affected by lumbosacral radiculopathy and the vast majority can be managed conservatively. If indicated, simple discectomy is the most common surgical treatment option. However, it is unclear if it prevents back or leg pain from persisting or returning beyond 2 years after the procedure.
The Barricaid device is first-of-its-kind, but since the surgery itself has unclear benefits, the device should only be approved if it adds a proven meaningful benefit for patients. There is evidence of benefit at 24 months post-op. However, the trial failed to demonstrate benefits in the long run, and raised many concerns of potential harm. Patients deserve better evidence.
We have several major concerns.
#1. If approved, Barricaid would be indicated for anular closure following limited discectomy. Yet, in the study, 62% of surgeries included carving out the anulus, a more extensive surgery. In other words, the proposed indication does not match up with the surgery for which the device was tested.
This raises important questions: 1) How effective is the device at closing natural anular defects without anulotomy? 2) Was anulotomy responsible for the increased reherniation seen in both Barricaid and control subjects?
#2. Long-term performance data do not indicate effectiveness. It is worrisome that the device had a somewhat superior outcome at 24 months post-op but not after that, because we already know that the benefits of surgery decrease after 2 years. Patients need a way to prevent their back pain and symptoms from coming back or a way to postpone the need for a second surgery. The long-term data do not demonstrate that the device meets either of these needs. The FDA’s analysis indicates that 50% of Barricaid subjects had reherniation compared to 70% control, however, the 20% advantage of Barricaid over surgery alone drops down to a nonsignificant 6% in the long run. In addition, although there was a decrease in second surgeries in the Barricaid group (9% vs. 16%) during the first 24 months, there were numerous second surgeries in the Barricaid patients that occurred after 24 months that should be counted as failures. And, the Barricaid patients had more second surgeries that required implanting hardware –a higher risk procedure.
#5. Our final concern is that the surgeons only used the 8 or 10mm device. There are no data on the 12mm size. Without data, the 12mm size should not be considered for approval.
In conclusion, patients who suffer with back pain need better alternatives for long term relief. The Barricaid device might be the answer, but the limited data provided suggests it is unlikely to offer the kinds of benefits that patients need. The study indicates substantial device failures. Its use in limited discectomy remains unknown, because most subjects had more extensive surgeries. In addition, the short-term benefits appear to be modest and wane after 24 months. Long term studies with blinding conducted inside the US may provide more certainty, but at this time the data are insufficient to recommend approval.
We respectfully urge you to vote NO because the data do not prove that the benefits outweigh the risks or provide a reasonable assurance of safety or effectiveness. Thank you for the opportunity to share our perspective.
The Medical Device Advisory Committee voted 12-1 (1 abstention) on efficacy, 5-9 on safety, and 5-8 (1 abstention) on whether benefits outweigh risks. Overall, the committee believed the Barricaid device worked, but most believed it does not have proven safety and the benefits don’t outweigh the risks. Read more about the meeting here.